Working Principle of a Mobile Metal Analyzer
The principle of the analysis method of SPECTRO´s portable and mobile metal analyzers is optical emission spectroscopy. Sample material is vaporized with the testing probe by an arc or spark discharge.
The atoms and ions contained in the atomic vapor are excited into emission of radiation. The radiation emitted is passed to the spectrometer optics via an optical fiber, where it is dispersed into its spectral components. From the range of wavelengths emitted by each element, the most suitable line for the application is measured by means
of a CCD.
The radiation intensity, which is proportional to the concentration of the
element in the sample, is recalculated internally from a stored set of
calibration curves and can be shown directly as percent concentration.
Download Schematics of the Mobile Metal Analyzer Working Principle
(pdf file, 40 KB)